Amprobe MO-100 Milliohm Meter features and accurate four terminal measurement down to 100µ ohms that does not require test leads resistance compensation. This milliohm meter indicates measurement errors due to temperature or connection problems.
|Chracteristics:||Amprobe MO-100 Precision Milliohm Meter|
|4 Terminal measurement down to 100 Micro ohms without the need for test leads resistance compensation|
|Measures from 200.0m ohms to 2000 ohms with a 100 Micro Ohm minimum resolution|
|Accurate measurement with three test currents coupled with over-temperature protection|
|Current lead resistance and potential lead resistance checks|
|Automatic Hold feature|
|Automatic Shut-off Feature|
|Compact and Lightweight|
|Notifies measurements errors due to connection or temperature problems|
|Specifications:||Amprobe MO-100 Milliohm Meter|
|Measuring Ranges||0-200.0m ohms in steps of 100µ ohm 0-2000m ohms in steps of 1m ohm 0-20.00 ohms in steps of 10m ohm 0-200.0 ohms in steps of 100m ohm 0-2000 ohms in steps of 1 ohm|
|Accuracy||±0.5% of reading ±2 digits over the operating temperature range, -15°C to 55°C (5°F to 131°F), with the supplied test leads|
|Test Current||1mA =>2000 ohms range 10mA =>200/ 20 ohms ranges 100mA =>2000m / 200m ohm ranges|
|Protection Fuses||Mains = 0.5A, HBC, 5 x 20mm, DIN Current = 0.5A, HBC, 5 x 20mm, DIN Voltage = 0.5A, HBC, 5 x 20mm, DIN Maximum Output Voltage (C1-C2) 20V|
|Dimensions||250 x 190 x 110mm (9.8 x 7.5 x 4.3 in)|
|Weight||1500 g (3.3 lb)|
Proposition 65 Warning for California Residents: This product can expose you to chemicals including Benzene, which is known to the State of California to cause cancer and birth defects or other reproductive harm. For more information go to www.P65Warnings.ca.gov.
|Applications:||Amprobe MO-100 Milliohm Meter|
|Check windings opens or shorts in motors, generators and transformers|
|Check wire-to-wire connection resistances|
|Heating elements testing|
|Resistance in critical electrical connections|
This meter enabled us to isolate and identify a defective appliance cord that was causing nuisance CAFCI trips. Isolation did not work, failures were minimum 4 days apart.Moving the load stopped the failures (cord flexed differently). This is now a new weapon in our troubleshooting arsenal.
Very useful in the HVAC/R Trade.
I used this tool to prevent major calls in the middle of the night. Its helps me generate sales to customers. I show them that their compresors, motors, transformers are about to go out, and most of them want to change the bad part with in a day or two. Of course, I had customers who have wished to listen to me.